Clinical and programming pattern of patients with impending deep brain stimulation power failure: a retrospective chart review
© Mehanna et al.; licensee BioMed Central Ltd. 2014
Received: 31 July 2014
Accepted: 28 August 2014
Published: 20 November 2014
It is important to prevent complications of implanted pulse generators (IPG) depletion by replacing the IPG in time.
We reviewed the charts of all patients with deep brain stimulation treated movement disorders who were seen at our institution over a period of 6 months. Among these, we retained for analyses those who had undergone IPG replacement within the previous 3 years.
A total of 55 IPG replacements (from 38 patients) were reviewed. Electrodes were implanted in the subthalamic nucleus in all Parkinson’s disease patients, in the ventral intermedius nucleus of the thalamus in all essential tremor patients and in the globus pallidus interna in all dystonia patients. Replacements were preceded by a voltage increase due to worsened symptoms in 27.3% (15/55); 25.5% (14/55) had full IPG depletion or had too low IPG reserve to allow for any voltage adjustment; and 21.7% (12/55) did not get a needed voltage increase either for safety reasons (eg: concern for increase in falls with higher voltages) or because the surgery date for IPG replacement was close. Only 25.5% (14/55) remained clinically well-controlled prior to IPG replacement, all of whom had IPG longevity estimates available. Clinical deterioration was noted prior to IPG replacement in 100% of patients without available longevity estimates versus 61% of patients with available longevity estimates (p < 0.001).
Despite best efforts, clinical deterioration prior to IPG replacement can be seen frequently. Routine estimation of IPG life, along with symptom assessment at every follow-up visit may prevent it.
KeywordsDeep brain stimulation Estimate Depletion
Deep brain stimulation (DBS) has been approved by the United States Food and Drug Administration (FDA) for the treatment of Parkinson’s disease (PD), essential tremor (ET) and dystonia, and is undergoing trials for approval in psychiatric disorders such as depression and obsessive compulsive disorders.
DBS acts through delivering an electrical current in a specific target area of the brain, the target being different according to the disease being treated. This current can be modulated through modification of voltage, frequency and duration of each electrical pulse delivered. The delivered energy creates an electrical field of variable size and shape according to the parameters used for stimulation . The current is generated by an implantable pulse generator (IPG), a small pacemaker-like unit that is implanted under the skin, usually in the chest or less frequently in the abdomen. The current is then delivered through an extension wire and implanted electrode to the target located deep in the brain. Once the IPG is depleted, it has to be replaced so current can continue to be generated and delivered to the brain.
Because IPG depletion can result in worsening of neurological symptoms [2, 3], and sometimes lead to medical emergencies [4, 5], it is important to prevent it by replacing the IPG in time. This can be done by routinely estimating IPG life and assessing symptoms at every follow-up visit. For the Soletra model, estimate can be done by telephoning Medtronic technical support. Reported DBS settings are entered into a computer software which estimates the longevity of the IPG from the date of implant. A web based application developed by the University of Florida is also available . However, the estimation of IPG life is always an approximation because the actual IPG life depends upon multiple specific DBS treatment parameters, duration of stimulation, lead impedance, as well as many other factors that cannot easily be estimated . While increasing DBS settings correlates inversely with IPG longevity , guidelines provided by medical device companies to help approximate the longevity of an IPG cannot take into consideration all possible therapeutic combinations of DBS and are thus far from accurate, especially because DBS treatment parameters change over time with the evolution of the underlying disease. There are many more sources of error that may flaw longevity estimation, including but not limited to device-to-device variation, decreased supplied voltage with battery usage, battery chemistry, impedance fluctuation and battery self-discharge through quiescent current .
There is little information in the literature regarding practices when DBS batteries are about to be depleted of power. We reviewed all IPG replacements done at our institution over a period of 42 months to survey our practice patterns, including DBS programming modifications, when DBS batteries are nearing end of life.
We reviewed the charts of all patients with movement disorders who underwent DBS surgery and who were seen at the Center for Neurological Restoration at Cleveland Clinic between 06/01/2012 and 12/31/2012. We retained for analyses those who had undergone IPG replacement within the previous 3 years. All batteries replaced being of the Soletra model (Medtronic ® Minneapolis, MN, USA), we included all replacements that were done for an IPG voltage of 3.69 or less, or for a drop by 0.3 V or more in the previous 12 months, or for clinical worsening that improved after IPG replacement, or at the family’s insistence because the predicted lifespan was reached. We excluded the IPG replacements that were done for other reasons (e.g. replacing the non-depleted contralateral IPG to spare the patient another surgery within a year). The patient’s age, gender, diagnosis, duration of disease at time of IPG replacement, IPG model and voltage at the last visit before IPG replacement were noted. The presence of symptoms attributed at the time by the treating clinician to IPG depletion and programming changes to address these symptoms were also recorded, as well as whether the IPG longevity estimate was known at the last visit before replacement, obtained through the Medtronic helpline. This was a minimal risk study utilizing existing data through chart review and not requiring any direct patient evaluation for the purpose of the study. Data was de-identified, informed consent was waived and the study was submitted to the Cleveland Clinic Foundation Institutional Review Board who exempted it from review.
Age of battery when replaced
Battery voltage when replaced
In 33 of 55 (60%) of the replacements, the IPG longevity estimate was available. Twenty of these (61%) had motor deterioration before IPG replacement, 5 of which benefited from a voltage increase prior to IPG replacement. 6 replacements were done at the patient’s and family insistence based on previous estimates, despite good IPG reserve and the lack of symptoms of IPG depletion. In contrast, all the 22 replacements that did not have estimates available had symptoms of depletion before IPG replacement. The difference in the proportion of patients who experienced worsening of symptoms prior to IPG replacement between the group with estimates available (61%) versus the group without available estimates (100%) was statistically significant using a Chi square test (Chi-square = 11.35; p < 0.001).
In our retrospective review of 55 IPG replacements in 38 patients, the most frequent diagnosis was PD (76.3%), followed by dystonia (13.2%) then ET (10.5%). We have found that the average longevity of the Soletra IPG was consistent with previous reports for patients with dystonia at 1.9 years [9–12] and PD at 4.4 years [6, 8, 13, 14]. However, the IPG lasted longer that previous reports in our patients with ET (5.75 vs 2 to 4 years) [6, 8, 12, 13], likely because most of our ET patients turn their IPG off at night precisely to prolong its life. Overall, this was consistent with previously published data that the underlying diagnosis can also affect IPG longevity with the characteristics of the stimulation target as well as the disease pathophysiology likely contributing to battery longevity [6, 12, 13], with dystonia typically depleting the IPG faster than PD or ET.
Clinical deterioration prior to IPG replacement occurred frequently. Fifteen of 55 (27.3%) replacements benefited clinically from a voltage increase (mean voltage increase = 0.26 V; SD = 0.1; range: 0.1-0.5 V). Interestingly, side effects noted after IPG replacement often required a voltage re-adjustment to their pre-power depletion levels. In 14 of 55 (25.5%) replacements, patients were clinically worse but could not get a needed voltage increase because of either full IPG depletion or too low IPG reserves. Finally, 14 of 55 (25.5%) IPGs were replaced well ahead of full depletion and no programming adjustments were necessary before or after IPG replacement. All of the latter had IPG longevity estimates available. We believe that knowing these longevity estimates prevented clinical deterioration prior to IPG replacements in several patients in our cohort—clinical deterioration was noted in 100% of IPG replacement without longevity estimates, whereas it was seen in 61% of IPG replacements with available longevity estimates (p < 0.001).
Available data regarding the clinical usefulness of longevity estimate are conflicting. In a prospective study on 72 IPG replacements, Stewart and Eljamel  demonstrated large differences between the actual longevity and the longevity predicted using a computer-based estimate integrating current DBS parameters. There was no correlation between the 2, and the actual longevity was shorter by a year in some instances. However, in a retrospective review of 320 charts, Fakhar et al. demonstrated that an University of Florida web based estimator/smart phone application (r = .67, p < .001) as well as the Medtronic helpline estimate (r = 0.74, p < 0.001) were correlated with the actual IPG life .
Our study replicate the results from Fakhar et al., albeit on a smaller sample and without the use of the University of Florida estimator. These studies actually complement each other with an emphasis on the physical and electrical aspect in Fakhar et al.’s, and on the clinical aspect in ours. The IPG longevity in ET patients was, however, longer in our study (5.75 vs 3.54 years). In our study, knowledge of the estimate was associated with a lower rate of symptomatic IPG depletions. However, because only 4 of the 55 IPGs were totally depleted at the time of surgery, we could not compare the estimated lifetime to the actual lifetime of the IPGs.
Finally, all the patients in this study were found to have the Soletra model of IPG. This specific model is old and has been replaced by newer models such as Activa. The newer models have an elective replacement indicator message that is displayed on patient and clinician programing devices, and suggests that a replacement may be required within approximately three months. The newer models probably predict battery failure more accurately, and the results of our study might not apply to them. However, clinicians taking care of the non-negligible number of patients still carrying Soletra, and until these deplete and get replaced by a newer model, will find our results still useful and clinically relevant.
The patients who had optimal symptom control were those seen well in advance of power depletion, with enough time to have batteries replaced before their symptoms worsened. However, clinical deterioration just prior to end of IPG life was not uncommon. A significant percentage of patients required voltage adjustments prior to IPG placement to improve their worsening clinical state, and often required another voltage readjustment after IPG replacement. Because IPG depletion can lead to worsening of neurological symptoms and even medical emergencies on one hand, and overzealous IPG replacement may not be cost-effective on the other, DBS programming clinicians should routinely estimate IPG life and assess symptoms at every follow-up visit in order to decide when IPGs should be replaced.
Implanted pulse generators
Deep brain stimulation
Food and Drug Administration
- Mehanna R, Lai E: Deep brain stimulation in Parkinson’s disease. Transl Neurodegener 2013,18(2):22.View ArticleGoogle Scholar
- Okun MS, Tagliati M, Pourfar M, Fernandez HH, Rodriguez RL, Alterman RL, Foote KD: Management of referred deep brain stimulation failures: a retrospective analysis from 2 movement disorders centers. Arch Neurol 2005, 62: 1250–1255. 10.1001/archneur.62.8.noc40425View ArticlePubMedGoogle Scholar
- Vora AK, Ward H, Foote KD, Goodman WK, Okun MS: Rebound symptoms followingbattery depletion in the NIH OCD DBS cohort: clinical and reimbursement issues. Brain Stimul 2012, 5: 599–604. 10.1016/j.brs.2011.10.004View ArticlePubMedGoogle Scholar
- Hariz M, Johansson F: Hardware failure in parkinsonian patients with chronic subthalamic nucleus stimulation is a medical emergency. Mov Disord 2001, 16: 166–168. 10.1002/1531-8257(200101)16:1<166::AID-MDS1031>3.0.CO;2-SView ArticlePubMedGoogle Scholar
- Alesch F: Sudden failure of dual channel pulse generators. Mov Disord 2005, 20: 64–66. 10.1002/mds.20354View ArticlePubMedGoogle Scholar
- Fahar K, Hastings E, Butson CR, Foote KD, Zeilman P, Okun MS: Management of deep brain stimulator IPG failure: IPG estimators, charge density, and importance of clinical symptoms. PLoS One 2013, 8: e58665. 10.1371/journal.pone.0058665View ArticleGoogle Scholar
- Montuno MA, Kohner AB, Foote KD, Okun MS: An algorithm for management of deep brain stimulation IPG replacements: devising a web-based IPG estimator and clinical symptom approach. Neuromodulation 2013, 16: 147–153. 10.1111/j.1525-1403.2012.00457.xView ArticlePubMedGoogle Scholar
- Ondo W, Meilak C, Vuong K: Predictors of IPG life for the activa soletra 7426 neurostimulator. Parkinsonism Relat Disord 2007, 13: 240–242. 10.1016/j.parkreldis.2006.11.002View ArticlePubMedGoogle Scholar
- Isaias IU, Alterman RL, Tagliati M: Deep brain stimulation for primary generalized dystonia: long-term outcomes. Arch Neurol 2009, 66: 465–470.View ArticlePubMedGoogle Scholar
- Blahak C, Capelle HH, Baezner H, Kinfe TM, Hennerici MG, Krauss JK: IPG lifetime in pallidal deep brain stimulation for dystonia. Eur J Neurol 2011, 18: 872–875. 10.1111/j.1468-1331.2010.03290.xView ArticlePubMedGoogle Scholar
- Lumsden DE, Kaminska M, Tustin K, Gimeno H, Baker L, Ashkan K, Selway R, Lin JP: IPG life following pallidal deep brain stimulation (DBS) in children and young people with severe primary and secondary dystonia. Childs Nerv Syst 2012, 28: 1091–1097. 10.1007/s00381-012-1728-6View ArticlePubMedGoogle Scholar
- Rawal PV, Almeida L, Smelser LB, Huang H, Guthrie BL, Walker HC: Shorter pulse generator longevity and more frequent stimulator adjustments with pallidal DBS for dystonia versus other movement disorders. Brain Stimul 2014, 7: 345–349. 10.1016/j.brs.2014.01.008View ArticlePubMedPubMed CentralGoogle Scholar
- Bin-Mahfoodh M, Hamani C, Sime E, Lozano AM: Longevity of batteries in internal pulse generators used for deep brain stimulation. Stereotact Funct Neurosurg 2003, 80: 56–60. 10.1159/000075161View ArticlePubMedGoogle Scholar
- Halpern CH, McGill KR, Baltuch GH, Jaggi JL: Longevity analysis of currently available deep brain stimulation devices. Stereotact Funct Neurosurg 2011, 89: 1–5. 10.1159/000321710View ArticlePubMedGoogle Scholar
- Stewart CD, Eljamel S: Prediction of implantable pulse generator longevity in deep brain stimulation: limitations and possible solutions in clinical practice. Stereotact Funct Neurosurg 2011, 89: 299–304.PubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.